Orifice plate Taps

Why Taps are required?

The area of the fluid stream continues to contract after the stream has left the orifice and it has a minimum diameter at the vena contracts. The pressure of the fluid, therefore, continues to fall after leaving the orifice. There is a slight fall in pressure in the approach section and the static pressure is at a minimum about one pipe diameter before the orifice plate. The pressure of the fluid then rises near the face of the orifice. There is then a sudden fall of pressure as the fluid passes through the orifice, but the minimum pressure is not attained until the venacontracta is reached.

Beyond the vena contracts, there is a rapid recovery in the static pressure. Owing to friction and dissipation of energy in turbulence, the maximum downstream pressure is always lesser than the upstream pressure. The pressure loss so caused depends upon the differential pressure and increases as the orifice ratio decreases for a given rate of flow. The differential pressure obtained with an orifice plate will depend upon the position of the pressure taps. The points to be observed while locating the taps are :

• They are in the same position relative to the plane of the orifice for all pipe sizes.

• The tap is located at a position for which the slope of the pressure profile is at least, so that slight errors in tap position will have less effort on the value of the observed pressure.

• The tap location in the installation is identical with that used in the evaluation of the coefficients on which the calculation is based.

There are five common locations for the differential pressure taps :

• Flange taps

• Venacontracta taps

• Full flow or pipe taps and

• Corner taps.

Flange taps:

They are predominantly used for pipe sizes 50 mm and larger and the centerlines are 25 mm from the orifice plate surface. They cannot be used for pipe size of less than 35 mm diameter. Since the vena contracts may be closer than 25 mm from the orifice plate. Flange taps are the most popular tap location for orifice meter runs on large pipes. Flanges may be manufactured with tap holes pre-drilled and finished before the flange is even welded to the pipe, making this a very convenient pressure tap configuration. Most of the other tap configurations require drilling into the pipe after installation, which is not only labour-intensive but may possibly weaken the pipe at the locations of the tap holes.

Venacontracta taps:

Venacontracta taps offer the greatest differential pressure for any given flow rate, but require precise calculations to properly locate the downstream tap position. Radius taps are an approximation of venacontracta taps for large pipe sizes (one-half pipe diameter downstream for the low-pressure tap location). An unfortunate characteristic of both these taps is the requirement of drilling through the pipe wall. Not only does this weaken the pipe, but the practical necessity of drilling the tap holes in the installed location rather than in a controlled manufacturing environment means there is considerable room for installation error.